- Specimen Collection Swab
- CHG Swab & Applicator
- VTM Kit
- Saliva Collection Kit
- Transport Medium
- Plastic Cryo Tube
- Face Mask
- Featured Products
Model Number :MFS-95000GJJ
Material：Flocked Head+ABS Handle
The product is suitable for collecting Gynecological HPV specimens from the vagina, genitourinary tract, etc.
Gynecological swab MFS-95000GJJ is a flocked swab of sampling for Endocervical clinical diagnostic, it utilizes state of the art “spray on technology” that the flocking process by means of an electro-static charge perpendicularly attaches millions of nylon microfibers on the medical grade handle tip.
Appearance: The appearance of the swab is neat, smooth, uniform in color, free of burrs, mildew spots, scars, scratches, cracks and other defects. Clean, odorless and tasteless. It feels soft, no yellow spots, stains, or foreign objects.
Performance: The pull-out force at the junction of the connecting rod and the sampling head is not less than 2N.
The swab can withstand a static pressure of 4N perpendicular to the axial direction for 15s without permanent deformation or breaking. The breaking force of the depression at the sampling tip is not less than 2N.
① If necessary, use a swab to wipe off the excessive secretions of the cervix, and then insert the disposable sampling swab into the cervix, and gently rotate the swab clockwise 3~5 times
② Slowly pull out the disposable sampling swab, put it into the sampling tube containing the cell preservation solution, break off the tail of the excess swab at the mouth of the tube, and leave the swab head in the sampling tube.
③ Tighten, fully immerse the tip of the flocking in the vial, and tighten the cap tightly, record the patient’s name and ID number on the label and paste it on the vial, put the vial in the specimen bag and transport it to laboratory.
Note: Too many rotations will cause dilution or loss of the sample.
After sampling, make sure that the disposable sampling swab does not touch any other objects.